DIN 53516 DIN Abrasion Tester DW5440
DIN Abrasion Tester, also called DIN Rubber Abrasion Resistance Tester, DIN Abrader or Rotary Drum Abrader, is used to determine the abrasion resistance of polyester soles, shoe soles and polymer sheet material.
FYI is a leading manufacturer and supplier of high quality DIN abrasion tester. We offer cost effective testing solutions and equipments for various industrial applications.
DIN Abrasion Tester DW5440
We provide reasonable price and high quality DIN abrasion tester. All the testing instruments manufactured by FYI factory are well-known throughout China and more than 50 countries abroad for providing superior value and performance.
DIN abrasion tester is applicable to flexible materials, rubber, tires, conveyor system, transmission belts, sole, leather and so on. There is adequate test area, operation and sampling are simple. It should be used together with a balance. It mainly consists of power system, rotary roller, sample clamping device, automatic stop system and rack and small driving gear for driving the sample, seat and dust collector.
The major working principle is that under some specific load and on sand cloth of a specific grade, the column-shaped sample is placed on the sand cloth for abrasion for a specific journey. In this way, the mass of the abrasion from the sample can be measured by using the balance. With this and the density of the sample, the abrasion volume can be calculated.
Follows are features of DIN abrasion tester:
- LCD display with blue backlight;
- Settable test cycles;
- Easy to operate
Application of DIN abrasion tester
DIN abrasion tester is suitable for the abrasion resistance test of polyester soles, shoe soles and polymer sheet materials. When the abrasion wheel rotates, the sample wears on the abrasive sheet of certain abrasive grade to evaluate its wear performance. The tester features good reproducibility and easy operation.
During the test, the sample and the grinding wheel are subjected to a certain Angle of tilt and a certain load, and the abrasion of the sample is measured in a certain range.
△V = (W1- W2) x 200 / (Q (S)
W1: pre-grinding weight (mg)
W2: grinding weight (mg)
The abrasion of the emery cloth should be between 200 ±20mg. If it is too sharp, please rough it with coarse grinding and correct it. When the abrasion is insufficient, please change the sand cloth so as not to affect the accuracy of the test.
DIN abrasion tester complies with ASTM D5963, DIN 53516, EN ISO 20344, GB/T 9867, GB/T 20991, GB/T 26703 and ISO 4649.
If you need more information about DIN abrasion tester, please contact us.
Standards of DIN abrasion tester
|ASTM D5963||DIN 53516||EN ISO 20344||GB/T 9867||GB/T 20991||GB/T 26703||ISO 4649|
Key specification of DIN abrasion tester
|Diameter of roller||150mm|
|Rotating speed of roller||40±1rpm|
|Horizontal speed of clamp||4.2±0.06mm per cycle|
|Load||2.5N, 5N, 10N|
|Power supply||AC220V, 50Hz|